Used to determine the level of SO2 in your wine. Contains 10 tests in the form of glass titrets. They can be used by themselves or you can purchase BMW510B which is a titret holder. Please keep in mind that due to some chemical interference, this kit is only accurate to within +/- 30ppm when used to test red wines.
Sulfite in Wine Titrets Kit
Range: 10-100 ppm as SO2
MDL: 10 ppm
Method: Ripper (Sulfite in Wine)
Increments: 10, 11, 12, 13, 14, 15, 16, 18, 20, 25, 30, 35, 40, 50, 70, 100 ppm
Kit comes in a cardboard box and contains everything needed to perform 10 tests: ten ampoules, ten valve assemblies, and instructions.
Results for this test kit are acceptable for white wines (although they can have an error up to 10 ppm). This kit is not recommended for use with red wines or white wines containing ascorbic acid or tannin. These wines often give false high test results.
Features of Titrets Technology
Dependable, sealed single-use titrants
Fast, easy procedure -- no need to count drops
No burettes or other equipment necessary
Perform a titration ANYWHERE
Titrets testers for monitoring Free SO2 levels in wines. The Sulfite in Wine Titrets test is based on the Ripper chemistry which employs an iodide-iodate titrant in an acid solution and a starch indicator2,3,4. The Ripper chemistry is used as a screening method throughout the winemaking industry for determining the sulfite content in wines. Results are expressed as free SO2. Results for this test kit are acceptable for dry white wines although they can have an error of up to 10 ppm. This test kit is not recommended for use with red wines or white wines containing ascorbic acid or tannin. These wines often give false high test results. As a rule, a test result of greater than 40 ppm free sulfite for any wine should be considered suspect and an alternative sulfite determination method should be employed.
Pack of 10 test vials and plastic tubes.
1. Collect the sample in any clean container.
2. Push a valve assembly onto the Titret ampoule tip until it fits snugly. NOTE: The valve assembly should reach the reference line on the neck of the ampoule.
3. Gently snap the tip of the ampoule at the score mark.
4. With the tip of the valve assembly immersed in the sample, squeeze the bead valve briefly to add a small amount of sample to the Titret. The contents will turn a DEEP BLUE color. Wait 30 seconds. NOTE: Do not squeeze the bead valve unless the tip of the valve assembly is immersed in the sample.
5. Rock the Titret to mix the contents.
6. Continue to add small amounts of sample until the liquid in the Titret turns from BLUE TO COLORLESS (or the color of your sample). Be sure to rock the Titret to mix the contents after each addition of sample. When the color of the liquid in the Titret changes to COLORLESS (or the color of your sample), the endpoint has been reached. Stop the test, hold the Titret with its tip pointed upward and read the scale opposite the liquid level to obtain the test results in ppm (mg/L) free SO2.